Difference between revisions of "RadiMation Error: 805309372"
Latest revision as of 08:24, 9 April 2021
Measuring at 150 kHz in a sweep with an activated "preselector" is not possible. Create a band below 150 kHz and a band above 150 kHz.
The R&S ESI analyser won't allow to set a start frequency below 150 kHz and end frequency at 150 kHz or higher with the preselector enabled.
- Create a multiband test with a band to max 149 kHz and one band from 150 kHz and higher.
- When not using a multiband test, then create two tests. Where the first test stops at 150 kHz, and the second test starts at 150 kHz.