The goal of sequence testing is to run different (type of) tests after each other, without the intervention of a test engineer. These tests can be of the same type or of completely different types.
The sequence testing functionality can for example be used to:
- Run a radiated emission test, a conducted emission test and a radiated immunity test (all performed in the same anechoic chamber) after each other.
- Another example is the situation where different amplifiers are used for different frequency bands. For this situation, multiple TSF files can be generated where each TSF file defines one of the frequency bands. All the created TSF files, can then be placed in a sequence so they can be run like one test. The same is true for antenna polarisation, where a horizontal and a vertical TSF file can be created and be put into a single sequence.
- three (or more) different conducted emission frequency bands can be added to a sequence file, and be tested sequentially after each other.
Especially in combination with the Pause test, sequence testing can be very attractive to easily automate a fixed combination of tests.
Creating and saving a sequence
From the main menu, select:
A new sequence file will be created. To add a test to a sequence, open the test configuration menu for the desired test, and select the desired TSF file.
This sequence file can be saved through the pull down menu:
Enter a description for this sequence file
select the desired TSF file from the list.
The TSF file will now be added in the sequence window as shown the picture below.
To save the sequence file, select:
Check if the correct sequence name is mentioned.
Adding a test to a sequence
When a sequence file has to be modified, the file can be opened from the menu:
The list of current TSF files in the sequence will be displayed.
To add a new test to a sequence, select the desired test from the Tests, pull down menu.
A list of the available TSF files for this test method will be shown. Select the desired TSF file by double clicking on it.
The TSF file will be added to the sequence file.
Changing a test in a sequence
The configuration of a TSF that is already included in a sequence, can be changed by selecting the test in the sequence and pressing the Edit button.
The TSF window of the selected test will be shown, and it is possible to change any parameter. When all the required parameters are changed, pressing the Store button will update the configuration of the TSF file in the Sequence, and the TSF window will be closed.
Removing a test from a sequence
On the other hand, TSF files can be deleted from the sequence list, by selecting the test in the sequence and pressing the delete button.
Changing the order of the tests
The order off the tests in the sequence can be changed by selecting the desired test and click the up or down arrow in order to promote or demote this specific test.
Starting a sequence
A sequence can be started by selecting from the menu:
A file selection window will be shown, in which a sequence file (*.SEQ) can be selected. As soon as the sequence file is opened, RadiMation® will start to perform all tests that are selected in the sequence, in the order in which they are specified.
A sequence overview window is shown with the tests to be executed.
Stopping a test in a sequence
There are two ways of stopping a test in a sequence.
- Stopping the currently running test in a sequence: To end the current test press the Stop button. RadiMation® will then interrupt the current test as soon as possible, save the already determined test data, and continue to the next test that was configured in the sequence.
- Stop sequence: There are two ways to stop a sequence, via the menu
Or via the Sequence overview window and use the button "Stop sequence". If this window is not visible use the menu,
A Pause test is available in RadiMation® to insert informational messages between two tests in a sequence. This Pause test can for example be used to direct the end-user to change the test setup, or it can be a reminder to turn off the amplifier. It is allowed to include multiple Pause tests (also after each other) in a sequence, so it can be used to easily automate a fixed sequence of steps.
The Pause test is available from the menu via:
The Pause Data window is used to configure the Pause test.
The following elements are available on this window:
|Note||Is the note (or informational message) that should be shown during the pause period, when the pause test is started.|
|Ok||Saves the current configuration of the pause test to the TSF file and closes the window.|
|Cancel||Closes the window, without storing the modifications to the TSF file.|
The informational message that was specified in the Pause Data window is shown in a bigger font, to make it more visible. The shown message can also be selected and copied to clipboard (using Ctrl+C).
If a timeout has been specified, the window will automatically close after the specified timeout period. It is however also possible to close the window with the Ok button.
The Pause window will remember its position and size, and will be shown on the same position with the same size, the next time it shown again. This allows the end-user to place the Pause window on a convenient and visible location.
The pause test will not store any test results, and therefore no test results will be added to the list of performed tests in the EUT file.